Electromagnetic flow converter for combination with OPTIFLUX and WATERFLUX flow sensors. KROHNE 03/ – – TD IFC R05 en. The documentation is only complete when used in combination with the relevant documentation for the . 2. The high-performance solution. IFC is the only electromagnetic flow converter with 3 x diagnostics of instrument, application and .
|Published (Last):||18 February 2006|
|PDF File Size:||20.95 Mb|
|ePub File Size:||13.31 Mb|
|Price:||Free* [*Free Regsitration Required]|
This warning refers to the immediate danger of burns caused by heat or hot surfaces. Function Of The Keys Please note: One converter for all applications, and thus advantages with regard to kronne, engineering and stock-keeping. Note the power supply krohnr and frequency Signal Cable A type DsConstruction Infrared interface for reading and writing all parameters with IR interface option without opening the housing. Only available, if pulse filter Fct.
Page 46 – Special measurements Page 47 Page 48 – Status messages and diagnostic informati Observe the national regulations for electrical installations! The ribbon cable of the display must not be folded or twisted repeatedly.
Detection of deposits or coating on the electrodes, temperature and conductivity changes in the medium, gas bubbles or solids, and an empty pipe are good examples of process diagnostics functions. Make sure not to damage the stranded drain wire 1 of the inner shield. Digital Flow Meter in Pune. S electronic temperature Upper limit for the permissible electronic temperature has been exceeded.
There are Replace electronic unit either defects or unknown krohnne. Ambient temperature Depending on the version and combination of outputs. Should you nevertheless need to return a device for inspection or repair, please pay strict attention to the following points: Or electrodes completely insulated krohhe.
Disregarding these instructions can result in damage to the device or to parts of the operator’s plant. The electronics plug-in module is identical, regardless of the housing version, providing the same functionality and a standardized operating concept.
Krohne ifc 300 Manuals
Construction of wall-mounted housing 1 2 3 4 krohnne 6 7 8 Cover for terminal compartments Terminal compartment for measuring sensor Terminal compartment for inputs and outputs Terminal compartment for power supply with safety cover shock-hazard protection Cable entry for signal cable Cable entry for field current cable Cable entry for inputs and outputs Cable entry for power supply 1 Turn lock to the right and open the cover. Starting The Signal Converter Protect housing from dust and water 1 2 3 4 Lay the cable in a loop just before the housing.
Calculation of mass flow with constant density of product; range: Symmetric pulse duty factor 1: S pipe not full Only for measuring sensors with 3 or 4 electrodes. F sensor global Data error in the global data of the measuring sensor electronic equipment. Ivc is the BTS cable.
KROHNE IFC Converter Quickstart EN User Manual | 68 pages
Connection diagram for measuring sensor, wall-mounted housing 1 2 3 4 5 6 Electrical terminal compartment in housing of the signal converter for signal and field current cable. F application error Application-dependent fault, but device is OK. Available inputs and outputs: Error messages possible via status output. Aluminium and aluminium sheet, partially polyester-coated Option Versions C and F: F sensor electronic Error, operational fault in sensor electronics.
We offer a wide variety of products to meet your unique needs. Check for the correct supply voltage printed on the nameplate. Measuring pipe not filled; function dependent on Fct. Special Measurement Tasks And Diagnostics Observe the following notes. Display And Operating Elements View Contact Call Seller Now.
The functions of the connection terminals depend on the settings refer to Function tables on lrohne The collection of personal data such as names, street addresses or e-mail addresses in the manufacturer’s documents is always on a voluntary basis whenever possible.